Wider blade opening than conventional Sugita Titanium Clips.
Non-clinical testing demonstrated that this clip is MR Conditional. In non-clinical testing, the image artifact caused by this clip extends approximately 5-mm from this clip when imaged using a gradient echo pulse sequence and a 3-Tesla MR system. (For more details, please refer to the IFU.)
Single Shaft Applier
For superior visibility and performance.
・Ultra low profile allows optimal visibility of operative site
・Support precise placement of Sugita titanium Aneurysm Clip II(T2)
・Meticulously milled jaw grooves help secure large and mini clip sizes
MR safety information
Non-clinical testing demonstrated that this clip is MR Conditional. A patient with this clip can be scanned safely in an MR system under the following conditions:
• Static magnetic field of 1.5 Tesla or 3 Tesla, only.
• Maximum spatial gradient field of 4,000 Gauss/cm (40 Tesla/m)
• Maximum MR system reported, whole body averaged specific absorption rate (SAR) of 2 W/kg for 15 minutes of scanning (i.e. per pulse sequence) in the Normal Operating Mode)
Under the scan conditions defined, this clip is expected to produce a maximum temperature rise of 2.2°C after 15 minutes of continuous scanning (i.e. per pulse sequence)
In non-clinical testing, the image artifact caused by the clip extends approximately 5 mm from this clip when imaged using a gradient echo pulse sequence and 3 Tesla MR system.
NOTE: Do not take the Sugita Clip Applier into the MR environment. This device contains material which has been determined to be MR Unsafe.